CLICK HERE to view the assembly diagram of the Passive Sampler.
CLICK HERE to view the Set Up Diagrams for Outdoor use of the Passive Sampler.
After assembly, the loaded sampler is placed into the re-sealable plastic bag, the bagged sampler is placed into the brown airtight container provided and taken to the exposure site.
After exposure, the above procedure is again followed to safely seal the exposed sampler in the brown airtight vial. Once indoors and in a clean area, the exposed pad is placed in a shipping vial (PS-118). It is then taken or sent to a laboratory for analysis.
Once the sampler is loaded and placed in its airtight vial, it should be kept at standard room temperature, as long as it is to be taken on site within several days.
If the loaded sampler is not going to be used for several days, it is best to place it in a refrigerator (not a freezer). It should be removed from the refrigerator about 12 hours before exposure for the purpose of equilibrating to the room temperature. This is done to prevent condensation from forming on the pad. Moisture is the enemy of the coated pad, as it can remove some of the collecting material.
CLICK HERE to view the configuration of the Passive Sampler for various gases.
It is a pre-coated pad, from the same batch, that is loaded into the sampler and put in its brown airtight vial to be used as a ‘control’. This ‘control’ sampler should accompany the samplers that are to be exposed, so it is subject to the same temperatures and conditions as the others, but the control sampler should never be opened, except to remove the pads and placed into an airtight shipping vial.
If pads are sealed in original glass vial within the aluminum vacuum pack – 90 days in refrigerator or 1 year in freezer.
If pads are opened and unused ones resealed in vial – 90 days in refrigerator.
Loaded sampler in bag placed in brown vial with lid – 60 days in refrigerator.
Extracted solution in water, in sealed 8 ml vial – 90 days in refrigerator.
Exposed loaded sampler, in bag placed in brown vial with lid – No refrigeration necessary as long as stored at normal room temperature, for no more than 14-21 days. The control pad must also be stored with exposed pads to detect any interference pads may accumulate during storage.
Note: Analysis should be done as soon as possible after exposure.
Not really, just the standard household refrigerator temperature. Note: Pads should be removed from the refrigerator about 12 hours before exposure for the purpose of equilibrating to the room temperatures. This is done to prevent condensation from forming on the pad. Moisture is the enemy of the coated pad, as it can remove some of the collecting material.
The concentration of NOx cannot be obtained by just using one NOx pad.
NOx is a mixture of NO and NO2, so there is no alpha value for NOx.
We must use both NO2 and NOx pads simultaneously for sampling, and the two pads are then analyzed separately.
The concentrations obtained are then both used in the analytical formula shown in the protocol.
The calculation for NO is done by subtracting NO2 from NOx.
To determine the concentration of NOx, one must use one NO2 and one NOx pad on each side of same sampler.
You will need one PS-100 sampler for every site and every gas that your customer wants to sample.
You can sample NO, NOx, NO2 with one sampler. NOx goes on one side and NO2 on the other side and then subtract the two to get NO.
You will use another sampler for O3, as it requires two collection pads to be used at the same time, one on each side of the sampler. The two pads are analyzed together.
You will use another sampler to do NH3, using 2 pads, one on each side of the sampler. The two pads analyzed together are considered one analysis.
Once the pad is exposed, it loses much of its sensitivity and can be stored in a normal room temperature for no more than 14-21 days. Naturally, the sooner the exposed pad is taken to the lab for analysis, the better. The coated pads are not affected by normal temperatures but should not be put in direct sunlight or in heated areas.
O3 ANALYSIS uses two O3 pads that are analyzed together, and is considered one analysis cost. NO Analysis uses one NOx pad and one NO2 pad on either end of the sampler.
NOx = NO2+NO and NO = NOx-NO2.
Each side is a separate analysis cost.
CLICK HERE to view the diagram for analyzing the Passive Sampler.
CLICK HERE to view the chart for Reagents, Methods and Analytes.
CLICK HERE to view the calculation for concentration in ambient air.